Publisher : Nofima AS
International Standard Numbers
Printed : 978-82-8296-693-1
Publication type : Nofima’s reports
Series : Nofima rapportserie 30/2021
Year : 2021
If you have questions about the publication, you may contact Nofima’s Chief Librarian.
The Karl Fischer Titrator ‘DL38’ from the producer Metler Toledo has been in use for several years, but it can no longer be repaired because of a lack of spare parts. After av validation with test of accuracy and precision the new Karl Fischer Titrator ‘915 KF Ti-Touch’ from Metrohm is approved as a replacement for the older ‘DL38’. The new Titrator is approved for accredited analyses in the method ‘BIOLAB A 13 Vann i Olje, Karl Fischer’.