Validation of Evolution 220
Publisher : Nofima AS
International Standard Numbers
Printed : 978-82-8296-646-7
Publication type : Nofima’s reports
Series : Nofima rapportserie 29/2020
Year : 2020
FULLTEKST : nofimaas.sharepoint.com/:b:/s/...
ARKIV : hdl.handle.net/11250/2660620
If you have questions about the publication, you may contact Nofima’s Chief Librarian.
BioLab has purchased a new spectrophotometer to ensure the operational reliability of a critical instrument. The new instrument is of the same type as the existing one (Evolution 220). A validation has been conducted regarding accuracy and precision (certified calibration carousel). A paired t-test has been done comparing the old and new instrument by the analysis methods Total Phosphorus, Soluble Phosphorus and Tryptophan. The purchased spectrophotometer Evolution 220 is approved for analysis.